OZ-AIR MUV Mobile UV Sterilizer (MUVS) features 4 No. High Output UVC lamps (106µW/cm˛ UV intensity @ 1m each lamp) for infection control through total room irradiation with germicidal UV.
OZ-AIR MUVS UV units are designed for virtually any room that needs air and surface disinfection, including sterile areas, laboratories, unoccupied patient rooms, clean rooms and all other applications, like hotel rooms, where permanent fixtures are not practical. Each UV unit is equipped with casters for maximum portability, a 25’ power cord, on/off switch with delay, timer and carry handle. MUVS can decontaminate a surface of 1.8 m from the device, within10 min. In case of larger rooms or rooms with an irregular geometrics, multiple units can be used of disinfection cycles can be repeated. MRS UV Unit Dimensions - 12”L x 12”W x 24”H
GENERAL OPERATION OF OZ-AIR MUVS
·Put protective goggles on ·Position unit in center of area to be exposed to germicidal UV light ·Plug unit into 220-230 VAC / 50 Hz / 1 PE / 0.35 KW Power Source ·Set timer to the desired exposure time BEFORE pressing the Start switch ·Press the Start /power on/ switch & Leave the room. Disinfection cycle begins after 120 Sec (2min) and will continue for the preset time until the timer turns the unit off. ·Keep the UV goggles with you in case you need to re-enter the room while the UV system is still in operation. Make sure that no one enters the room while UV is on. Lock the door(s) and place Do NOT Enter! signs. ·Reposition the unit to cover any shadowed areas, if needed, and repeat the procedure ·Check and clean the UV lamps regularly. Annual lamp replacement recommended. Avoid exposure to direct or reflected UV light.